Sinopharm - Oxygen Mask - High Concentration Mask
Non Technical and Terms
- Initial Delivery Time - 2 to 4 weeks
Pricing Incoterm - CIP
- Advance Payments - 40% upfront with a Bank guarantee, 60% on shipping
- Warranty - 12 months
- Manuals - Printed and digital user service manuals
- Shipping - Air, Land or Sea
All Training and Manual materials will be available in English. Some materials will be available in French and Spanish. Where French and Spanish are needed and not available, the supplier will offer translation services in electronic copy. Training shall be available in English or Spanish or French and shall be available to users within two (2) weeks of equipment delivery.
China National Pharmaceutical Group Co., Ltd. (Sinopharm) is a large healthcare group directly under the State-owned Assets Supervision and Administration Commission (SASAC) of the State Council, with 200,000 employees and a full chain in the industry covering R&D, manufacturing, logistics and distribution, retail chains, healthcare, engineering services, exhibitions and conferences, international business and financial services.
For further enquiries, please contact
Sinopharm Fortune International Trading Corp.
Address: Fortune Tower, No.4, Hui Xin Dongjie, Chao Yang District, Beijing 100029, China
Tel.: +86-10-8466 3076/ +86-13269815298
Also referred to as a non-rebreather mask. These masks can deliver up to a FiO2 of 1.00 (100% oxygen), but that is totally dependent on how well the mask is fitted to the patient. Typically, a FiO2 of approximately 0.80 can be expected with these masks. These masks only come in one size per patient population (adult/child). This mask is made from plastic and always has a reservoir bag attached to the bottom of it. The mask uses either 2 or 3 one-way valves. These valves prevent the patient from drawing in room air and also from exhaling back into the reservoir bag that contains pure oxygen from the source. This mask requires at least 12lpm of oxygen with 15+ lpm not being unusual. The amount of oxygen delivered to the mask must be sufficient so that the reservoir bag does not fully collapse on inspiration. These devices should be connected directly to the oxygen source and not to a humidifier. Oxygen tubing with a normal length of 2.1 meters is attached to the mask. Other lengths of tubing may be provided.