Meheco - Stethoscope (for a la carte vaccine products) HS-30A
Non Technical and Terms
Inspection and testing will be performed prior to and during production along with a final random inspection of finished products by a Bank approved independent inspection agency. The inspection contract will be let by Meheco to the approved inspection agency with costs passed through to the Borrower to pay.
Some countries have an offer from Meheco on a CIP freight basis, this is stated in the ROI. Where freight is offered, Meheco will organize the delivery of goods to the agreed destination and the borrower must pay for the carriage. The estimated freight costs are on each borrower’s ROI.
Where freight costs are not stated in a Borrower’s ROI then the carriage is Free on Board (FOB). The borrower is responsible for the cost of shipping and insurance from the country of origin to the product’s final destination in their country.
Note: Regardless of the above some countries may wish to collect the products, which can be accommodated on request.
30 days, as above.
40% on order; 60% on final acceptance of product
China Meheco Corporation is a State-Owned Enterprise in China. It is an established large-scale medical distributor with a wide distribution network. It has access to many China based manufacturers with large production capacity and some stock available for faster delivery.
Contact - For technical questions or assistance determining your requirements contact: Wenhui Sun, Project Director, Meheco Corporation. Email: 孙文辉 firstname.lastname@example.org
Purpose: The purpose of these devices is for listening to sounds from the heart, lungs, and/or gastrointestinal tract.
Description: A stethoscopes comprises of a membrane at the listening head connected by a split "Y" tube to the headgear with ear olives that are placed into the users’ ears. The intended purpose for this situation is for the monitoring of a blood pressure. Specifications:
1. Device shall be sensitive to sounds from 50-500Hz.
2. Diameter of tubing shall be approximately 10mm in diameter.
3. May be disposable/single patient use or reusable.
4. If reusable, all materials shall have the ability to withstand commonly use disinfectant wipes.
5. All material utilized must be latex free.
6. The device may utilize either a single or dual head.
7. The length of the device shall be from 50cm – 61 cm.